CENTRAL STERILE SUPPLY DEPARTMENT
The main objective of CSSD is to ensure supply of sterilized material to the hospital. The department is located on Level-II, Cobalt Block, Nehru Hospital and works 24 x 7 throughout the year.
Functions of CSSD
1. Provides supply of sterile instrument sets and all other sterile material used in patient care.
2. Maintain effectivness of cleansing, disinfection and sterilization processes.
3. Monitor and ensure controls necessary to prevent cross infections.
4. Maintain supplies and equipment inventory.
5. Stay updated with development within the field.
6. The department also provides training to B.Sc. O.T. students, nursing students and sponsored candidates from other hospitals under in service programs.
Ministerial Staff 1) Storekeeper 2) LDC
Group C Staff (Erstwhile Group D Employees) 1) Hospital Attendant
2) Sanitary Attendant
The CSSD is divided into four functional Sections:
1. Procedure Sets Section
2. Dressing Section
3. Vials Section
4. Autoclaving Section
I) Procedure Set Section
This section provides procedure set services to the entire hospital including wards and OPD's, following sets are prepared in the section.
a) Cut Down Set
b) Catheter Set
c) Lumbar Puncture Set
d) Suture Set
e) Tracheostomy Set
f) Delivery Set
g) P V Examination Set
h) Haemodialysis Set
i) Dressing Pack Set
II) Dressing Section
Following types of Dressings are prepared in this section :-
a) Small Dressing
b) Burn Packs
c) Burn Gathri
d) Gynae Pad Pack
e) Gauze used in operation theatre
III) Vial Section
Following type of vacutainers are distributed to the entire hospital:-
e) PTI (Coagulagram)
IV) Autoclaving Section
This section provides service of sterilization of instruments/ materials required in the hospital. Sterilization is done by two ways:-
1) Steam Sterilization
2) E.T.O. Gas Sterilization
To ensure that appropriate sterilization cycle parameters have been achieved. Sterilization is monitored by following ways:-
a) Chemical Indicators
b) Biological Indicators
c) Mechanical Indicators
d) Bowie Dick Type Test
CHEMICAL INDICATORS :-
These are monitoring devices that respond with a characteristic chemical/ physical change to one or more physical conditions in sterilizing chamber. They are intended to detect potential sterilization failures that may result from personal errors or sterilization error. These usually consists of sensitive chemical or ink that changes colour in specificed parameters. At the time of completion of cycle, the indicators are examined for appropriate colour change.
=> ETO Indicator Tape ( Turns from Yellow to Copper Red ), Steam Indicator Tape ( Turns from Yellow to Black)
=> Class VI Indicator ( Turns from Blue to Green)
BIOLOGICAL INDICATORS :-
These are calibrated population of bacterial spores (of high resistance to the mode of sterilization being monitored) in a carrier, put up in a package which maintains the integrity of inoculated carrier. These are used to test the ability of a sterilization cycle to kill microorganisms. It is done on weekly basis for steam sterilization and with every batch for ETO Sterilization.
B.I. used for steam sterilization contains spores of Bacillus Stearothermophillus.
B.I. used for E.T.O. Sterilization is Bacillus Atrophaeus.
MECHANICAL INDICATORS :-
Physical/ Mechanical Indicators monitors include real time records of chamber temperature and pressure, displayed on various types of recording chartd, computer printouts and gauges. Mechanical monitoring is useful for the early detection of malfunctions, so that alternative procedure can be used in the event of sterilization failures. Mechanical monitors must be checked before, during and after each sterilization cycle to verify that appropriate parameters for sterilization have been reached and maintained for the correct exposure time. The recording chart or computer printout is signed by the operator. These charts and printouts are maintained with sterilizer records.
BOWIE DICK TYPE TEST :-
This test is performed weekly. The objective of Bowie Dick Type Test is to assure that the air removal device is working properly i.e. air is completely removed from the chamber and during sterilization process if any air leak occurs it could be detected by this test.
----------------Before Sterilization --------------------- After Sterilization--------------
HUMAN RESOURCE DEPLOYMENT
51st Annual Report (2017 – 2018)
NEHRU HOSPITAL CSSD
NEHRU HOSPITAL CSSD
CSSD at Cardiac Centre Which cater the Advance Paediatric Centre and Advance Cardiac Centre.
All the procedure and services are same as at Nehru Hospital, CSSD.
52nd Annual Report (2018 – 2019)
COMPARATIVE STATEMENT FOR LAST THREE YEARS
Advance Cardiac Centre, CSSD